Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
Data Zero Manual Null Buttons
The Null buttons are used in Data Zero Manual mode to set the true zero position on a DC-coupled measurement system. When you press the button, the Manual Adjust value is automatically set to the current average value for that channel, as determined by averaging 1024 samples.
To use the Null buttons, you first set the external measuring system to produce an effective zero input condition on a channel. This may involve venting a relative pressure sensor to atmosphere, putting a thermocouple into an ice bath (or into thermal contact with the reference temperature), or shorting the input leads of a voltmeter circuit. When you have done this, whatever DC level is read by the ADC must be treated as the effective zero, even if it is not zero volts. Hitting the Null button for that channel will do this automatically.
LI.ZeroNull=1 nulls the Left Input channel. The null operation takes place no matter what value you supply in the macro, including 0.
Similarly, RI.ZeroNull sets the Right Input, LO.ZeroNull sets the Left Output, and RO.ZeroNull sets the Right Output.
Alternatively, you can use the Ch. prefix to set a channel previously specified via the Ch Channel Select command.
Note that attempts to use these commands when the controls are disabled in Spectrum, Spectrogram, or Pitch Track modes will result in "Control disabled. Aborting macro."
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 30 Years of Innovative Instrumentation
© Copyright 2007 - 2017 by Interstellar Research
All rights reserved