Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
Main Tone Frequency
Controls the frequency of the waveform for the stream.
The maximum frequency resolution is limited by the effective sample rate:
Resolution (Hz) = Sample Rate / (65536 * 8192)
For a sample rate of 48000 Hz, the resolution is thus 0.000089 Hz.
The actual frequency entry process is controlled by the Step Mode shown on the button below the frequency control. This defaults to Step Direct. If you click on this button to open the Step Mode dialog, and enter 0 in the value under Step Direct (or under Step Hertz if selected), the step value will show as 0.00001 Hz. Daqarta interprets this to allow scrolling of the Tone Frequency at the finest (89 ppm) resolution.
You may want 6 or more decimal places for this. By default, all Generator frequency controls show 5 digits after the decimal. You can adjust this between 0 and 9 using the Frequency Places control in the Step Mode dialog.
When Play Wave type is active, the Freq control changes to Rate. The Play Rate control allows Rate to be set relative to the absolute rate (Abs), as recorded originally, or relative to the current sample rate (Rel).
Note that when Freq is set to 0, the output is a constant level controlled by the Phase setting; at a Phase of 90 degrees the output of a Sine wave is a constant 100%, for example. This can be useful when used with Burst or Phase Modulation to modulate another stream.
L.1.ToneFreq=1k sets Left Stream 1 Tone Frequency to 1000 hertz.
L.1.ToneFreq=>1 increments the frequency to the next higher Frequency Entry Step, while L.1.ToneFreq=>-1 decrements to the next lower step. Only +/-1 step is accepted.
Note that the ToneFreq command will change the indicated frequency value even if that stream is using a Noise-type Wave that doesn't allow normal user access to the Frequency dialog. In that case, there will be no change to the output signal until a non-noise Wave type is selected.
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 30 Years of Innovative Instrumentation
© Copyright 2007 - 2013 by Interstellar Research
All rights reserved