Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
Frequency Counter - RPM Measurement
For use as an engine tachometer, this mode works in conjunction with the Cylinders control to compensate for the number of cylinders on an engine, where you are triggering on spark or fuel injection pulses. Be sure to read the Engine RPM Measurement topic under Automotive Applications in the Applications section. It contains information about Trigger settings, input voltage limiters to prevent damage to your sound card, and tips on how to obtain trigger signals from spark plug wires or fuel injectors without direct electrical contact.
For electric motors or other rotating equipment, set Cylinders to twice the number of pulses per revolution. The default setting is 2, which assumes one pulse per revolution.
Observe the same considerations as for Hertz Measurement with respect to triggering.
Note that these RPM measurements have high resolution (fractional RPM, if desired) even at 10 updates per second (Fast). Conventional benchtop frequency counters count the number of pulses in a one-second or ten-second interval, and require you to multiply the reading by 60 yourself to convert to RPM. Even after waiting ten seconds, you would still only get 6 Hz effective resolution.
FcountMode=RPM or FcountMode=1 sets the Frequency Counter to RPM mode. See Macro Notes under Frequency Counter - Hertz Measurement for the complete list of mode names and numbers.
Freq is an internal variable that holds the current Frequency Counter readout value, dependent on the chosen mode (Hz, RPM, msec, Total, or SpecTot). You can assign it to a variable such as F via F=Freq, or you can display it via Msg=Freq, or use it directly in an expression.
Total holds the count of total cycles counted since the last Reset, valid during Hz, RPM, or msec modes. This allows macros to get the Total value even when these other modes are active. Note that this does not work in SpecTot mode.
See also Frequency Counter
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 35 Years of Innovative Instrumentation
© Copyright 2007 - 2023 by Interstellar Research
All rights reserved