Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
SPL Response Decay
This controls the Decay time of the Sound Level Meter response. Values are in integer milliseconds only, so to enter, say, 1 second you would enter 1000.
The Decay time works with Attack as a "quasi-peak" detector. Decay controls how rapidly the meter reading dies down after a drop in sound level. Long Decay values make the meter easier to read when the input signal is rapidly changing.
If you set Decay to zero, Daqarta assumes you do not want quasi-peak operation. In this case, Attack sets a single time constant configured to provide an overall averaged response.
Decay has no effect if Attack is set to zero. In that case, the meter responds immediately to all changes, with no time constants.
SPL_TCdecay=1000 sets Attack to 1000 msec (1 second). SPL_TCdecay=>1 increments the current value and SPL_TCdecay=>-1 decrements it by 1 msec.
See also Sound Level Meter
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 35 Years of Innovative Instrumentation
© Copyright 2007 - 2021 by Interstellar Research
All rights reserved