Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
Sound Card Signal Generator Toggle
This toolbar button activates sound card output operations. You may also use ALT+G.
In order for the Generator to be successfully toggled on, either Left or Right (or both) outputs must be selected in the Generator Control Dialog. Otherwise, that dialog will open and 'Select Left and/or Right Generator channel' will appear on the message line.
When Generator and Input are active at the same time, it is referred to as "full-duplex" operation. Most modern sound cards are capable of this. If yours is not, then toggling Generator on will force Input off and disable it, and vice-versa.
Note that the Multi-Channel Outputs dialog button is disabled unless both Left and Right outputs are selected. Conversely, when the Multi-Channel Outputs On button is active in that dialog, even if the Generator is off so there is no actual output, the Left and Right output buttons in the main Generator dialog are disabled.
Gen=1 turns the Generator on, Gen=0 turns it off, and Gen=x toggles between the two states. These commands are always immediate.
If the On/Off Fade control in the Attenuator Options dialog is enabled for manual Generator toggles, Gen#F=1 toggles on with the specified fade-up, Gen#F=0 toggles off after a delay and fade-down, and Gen#F=x toggles between states using the fades.
See also Generator Introduction and Features
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 35 Years of Innovative Instrumentation
© Copyright 2007 - 2021 by Interstellar Research
All rights reserved