Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
Screen Zero Individual Adjust controls
These controls are disabled unless the Individual button is active. (They are also disabled in Spectrum, Spectrogram, or Pitch Track modes). When enabled, the positions of each waveform trace may be controlled individually.
Note: Units for this control are always percent, regardless of the Units button state.
Entering a positive value will move the trace down on the screen. For example, if you enter 25 the display will be shifted down such that a point that was formerly at 25% of full scale will now be at zero. Likewise, negative values will move the trace up accordingly.
Note that if screen magnification is changed (PgUp or PgDn keys), each trace will be magnified or reduced relative to its own Zero position.
If you have two bipolar traces and you want to give half of the screen to each, set the Individual control for the one you want on top to -50% and the one you want on the bottom to 50%.
For 4 bipolar traces, try these values:
LI -75 % RI -25 % LO 25 % RO 75 %
For 4 unipolar traces, try:
LI -50 % RI 0 % LO 50 % RO 100 %
LI.ZeroScrnAdj=25 will set the Left In screen zero to 25%. Similarly, RI. (Right In), LO. (Left Out) and RO. (Right Out) prefixes will set those waveform channels accordingly.
A macro can read the current Screen Zero setting of the Left In waveform channel via X=LI.ZeroScrnAdj.
Note that these controls are disabled unless the Individual button is active (depressed), and the display is in waveform mode (neither Spectrum, Spectrogram, nor Pitch Track active). If you attempt to use a ZeroScrnAdj command when the button is disabled, the macro will abort with a "Control disabled. Aborting macro." message.
You can check if this control is enabled via an E.IF.ZeroScrnAdj= control enable IF test. You don't need to specify which control via LI. or other prefixes, since all are enabled or disabled together.
See also Zero (Screen and Data)
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 35 Years of Innovative Instrumentation
© Copyright 2007 - 2023 by Interstellar Research
All rights reserved