Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
As noted under the Volume dB control section, Daqarta can obtain resolution much finer than the actual attenuator step sizes by making small adjustments using the effective Level settings for each Stream. The Use Level button selects that mode of operation. It is disabled unless Use dB is also selected, which is only possible after calibration.
When Use Level controls the effective Level, it does so transparently; there are no visible changes to actual Level control values. The effective value is computed as the product of the control setting times the appropriate factor needed for attenuator step interpolation.
So, why would you ever not want Use Level? If you want to change loudness with a minimum of clicks and pops. The reason is that when any parameter that affects waveform generation is changed, the sound is restarted with the new changes in effect. That applies to Level, even when it is done automatically via Use Level volume changes. Since the new sound doesn't just continue from where the old sound stopped, there will almost certainly be a discontinuity, which causes an audible click.
You can use the GenUpdate macro to avoid these clicks in certain situations, especially during macro control of volume or level.
If Use Level is not in effect, simple volume changes will only use the Wave and Master attenuators of the mixer. Those just operate on the analog output without affecting the computed wave, eliminating the need for a restart.
However, note that there may still be softer clicks depending on your sound card, driver, and even Windows itself. These are the same clicks you would hear when moving the Windows Mixer volume sliders for any sound on your system.
UseLevel=1 turns Use Level on, UseLevel=0 turns it off, and UseLevel=x toggles between on and off.
Note that this command is enabled only when Use dB is on. Otherwise, the macro may abort with an error message. To be safe, give UseCal=dB before UseLevel.
See also Generator Attenuator Options Dialog.
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 35 Years of Innovative Instrumentation
© Copyright 2007 - 2023 by Interstellar Research
All rights reserved