Data AcQuisition And Real-Time AnalysisScope - Spectrum - Spectrogram - Signal Generator
Software for Windows
Science with your Sound Card!
Contact us about
Trigger Normal Mode
Normal trigger mode, just like that mode on an oscilloscope, will wait forever for the signal to cross the specified trigger Level with the specified Slope. However, unlike many scopes, it will show 'TRIG WAIT' on the frozen prior trace if the wait is more than about one second. To break out of a wait for a trigger that doesn't arrive, you can either toggle Trigger off, or adjust your experiment, or change the trigger parameters. You could hit Auto or Auto Level to view the current signal, then set a new trigger Level based upon that.
TrigMode=Norm or TrigMode=2 sets the mode to Normal. See Macro Notes under Trigger Auto Mode for the complete list of mode names and numbers.
Questions? Comments? Contact us!We respond to ALL inquiries, typically within 24 hrs.
Over 35 Years of Innovative Instrumentation
© Copyright 2007 - 2023 by Interstellar Research
All rights reserved